WFCAM on-sky characterization tests
WFCAM ON-SKY CHARACTERIZATION TESTS
A. Adamson, P. Hirst (JAC)
M. Irwin, D. Evans, J.Lewis, S. Hodgkin (CASU)
Last update by PH 20031006 - add tweaks from JRL
This document lists the various on-sky tests which will be used to
characterise WFCAM as soon as possible after commissioning time, and
in some instances during commissioning.
I (PH) would like to use this document to define a list of what
observations are necessary and to some extent who will be responsible
for looking at the data and signing off each requirement.
DETECTOR NOISE PROPERTIES
The detectors will be being used in a new mains power environment,
attached to a telescope, and may have noise properties different to
those measured in the lab. The difference needs to be characterized so
that its impact on data reduction can be assessed. This will include:
- Basic read noise measurements (can be done in daytime)
- Re-measure the crosstalk matrix (could be done pre-final-alignment)
- Bad pixel stability (need bias and dark frames from the
earliest stages of commissioning)
- Dark current stability (daytime, to some extent)
- Repeatability and level of reset anomaly (daytime)
Basically check that it works. Verify 2x2, 3x3 and 4x4 sequences. Also finalise which option is optimal in the cases where more than one option is availiable for a given mode in terms of the step sizes (ie where there's more than one common denominator between the 2 pixel scales.
Note: in case of measurement of sky properties, the measurement
listed is to be carried out in every appropriate filter. Repeatability
of filter placement should also be tested, and this can be done in a
number of the tests below.
- Brightness, time variation, spatial scales of the emission.
- Effect of large spatial offset to a "sky" position.
- Sky brightness vs. distance from Moon (check for additional contribution
from scattered light).
- Sky brightness as function of zenith distance.
Can probably mostly be done with the sky measurement data above
- Fringe amplitude and stability during change of sky brightness.
- Variation time and spatial scales, speeds of motion.
Need an early measurement of the sensitivity limit in the standard
observing modes. Also should compare coadded images of the same field
taken on widely-separated nights. Quantify overheads in standard
Early check of background-limited exposure times.
Long exposure to pick up a lot of CRs - test whether DR rejects
them. Probably daytime to some extent. Can be done in conjuction with dark current monitoring.
- Check in a field with a range of point-source brightnesses.
- Measure the decay profile of a latent image bot in terms of time and number of reads and number of resets.
- Repeat at intervals to check temporal stability.
- Look for and quantify and adjacency effects.
- Dome flats (lamp on/off), fading twilight flats - assess
usability. Screen and illumination will be provided for this. (Will it? PH)
- Nonlinearity from dome flats.
- Vignetting function and its motion if any (ie dependence on
telescope attitude). Also includes taking a mesostep sequence.
- Assess colour dependence of flat field.
- Assess spatial thermal signature in the K band.
- Tests of different flat fielding algorithms.
- Assess flatfield stability (impacts on data taking and reduction
Observe a field with large numbers of stars, and one with a few
bright stars. Quantify the scattered light and ghosting.
- Place a UKIRT faint standard on each chip independently (if
possible, in each of the different channels on each chip).
- Observe a number of the predefined standard fields.
Dead-reckoning coordinate system - first go. Includes the following:
- Chip and readout orientation.
- Refine WCS constants.
- Refine radial distortion model (needed later for more accurate
- Use the same data to check microstepping accuracy.
- Test guiding in crowded fields.
- Test guiding in the field of a large, bright object.
- find the brightness of the faintest viable guide star as a function of sky conditions.
Require observations of standard fields as yet to be finalised